complete with decorative cabinetry, easy access panels, horizontal gas, communications and electrical consoles, and sized according to the detail drawings.
A completely assembled cabinet to be fitted to a surface mount headwall after rough-in is complete. Cabinetry to be of wood veneer over a 20 mm veneer core plywood frame, and to include two side cabinets, bifold doors with 270 hinges, decorative overhead valance with integral lighting, and lower safety panel. One matching centre panel to conceal all site connections. Wood veneer and stain to the architect's later selection.
One three foot. 2-lamp, fluorescent light, complete with up, down, and up/down lighting features.
Hinged overhead valance and hinged lower safety panel to cut off electrical supply to bed receptacle when contacted.
Heavy gauge 6063-T5 extruded aluminum alloy frame complete with clear satin anodized finish. Minimum 16 gauge satin-coat steel backing, suitable for mounting, Component assembled into a single, rigid wall mount structure by means of machine screws. All reinforcing and fastening devices to be concealed.
Gas, Communications and Electrical Compartment(s):
Aluminum or satin-coat steel barriers in backbox to separate electrical services, communications devices and medical gas outlets where required, Easy access finish fascia to include die-cut knockouts as required to mount all devices indicated on the drawings. Blank fascia knockouts to include an aluminum coverplate. Electrical power types to be separated by steel or aluminum barriers or EMT or flexible steel conduit where required.